|Place of Origin:||Hunan China (Mainland)||Brand Name:||SANDA ELECTRON||Model Number:||SD994AI-9K4Y|
|Display Type:||Digital Only||Measuring Range:||1~9999||Accuracy Class:||0.5|
|Operating Temperature:||-10~55℃||Dimensions:||96*96 & 91*91 (mm)||Certifications:||CE|
High Definition, Dot Array Screen Display
|Packaging Detail:||standard package|
|Delivery Detail:||Factory LT: 15 days|
High Definition,Dot Array Screen Display
suit grid,auto systems,MKH cabinet
LCD Ammeter, LCD Ampere Meter, LCD Current Meter
1. Product Features
1.1 True RMS measurement
1.2 High-definition, large screen, dot-matrix LCD displays 3 phase voltage or current
1.3 Optional the upper/ lower limit alarm function, I / 0 switch function
1.4 Programmable, it may be arbitrarily adjusted the displaying value between 1 -9999 by the keys on panel
1.5 It’s Optional for communication function, communication address and baud rate.
2. Technical Index
Voltage:500V (May customize for the special)
Current:5A, 1A (May customize for the special)
Lasting: 1.2 times
Instant: current: 2 times/second, voltage: 10 times/second
Extend measure range by instrument transformer or shunt
AC220V±10% (May customize for the special)
Display time for the data updating:1s
Isolation Withstand Voltage
Mean normal working time
Humidity:≤93% No corrosion gas
3. Basic operation and useage
3.1 Display & Operation
3.2 Menu structure
3.3 Home Menu Page
3.4The Device Menu of Setup
After entering the “SETUP”page, through the cursor movement enter the goal to set submenu, then set the
related parameter of the meter.
For example: Pressing the ENTER button to enter into “system parameter” to set submenu, then we may adjust
3 phase ammeter (TA Current Transformer ratio) or 3 phase voltmeter (TV Voltage Transformer Ratio) and the
mode of connection.
Note: mode of connection "Y" represents 3 phase 4 wire, "the Delta" represent 3 phase 3 wire as below:
As shown above: 3 phase ammeter CT: 100/5A
We should input 20A at the position of TA Ratio, then QUIT & SAVE.
4. Outline & Installing Dimension
My products are modularity structure, they may meet varies of demands.
We also provide customized serivce.
6.1 The following conditions will lead to the device be damage or work abnormal:
6.1.1 The auxiliary power supply, voltage, frequency are over-range.
6.1.2 Please do not treat the PT signal as well as the auxiliary power supply at the same time Unless the PT have
enough power, so that the meter can be ensured to work normally.
6.1.3 CT circuit current wiring terminal screws must be tightened, and ensure the inlet / outlet lines are reliable
to avoid faults.
6.1.4 If checking meter, the calibration instrumentation should be prior a class 0.1 magnitude, so that the
accuracy could be ensured.
Zhuzhou Sanda Electronic Manufacturing Co.,Ltd
Address: No.502, Wulidun Economic Development, Liling, Zhuzhou City, Hunan Province, China.