Small volume, open-room operation, free adjustment of film developing speed, sealed and environmentally friendly, the machine could automatically complete film development and fixation rinse. Therefore, the problems of environmental pollution and body injury to operating staff caused by manual operation of injection and darkroom have been totally resolved
New type with heating function
Multi-film developing disposal may be completed when the developer and fixer is at 18-25 centigrade.
One piece of film enters per 30 seconds for multi-fill developing disposal, and the development may last 2-8 minutes. It is adjusted in accordance with the exposure conditions of x-ray machine.
Multiple films of 2x3, 3x4, and 2.7x5.4 cm are applicable for the treatment of development.
Regular maintainance is convenient. No extra tool is needed to replace chemical solutions or wash relative parts.1. Introduction
The oral automatic X-ray film bath equipment is a total automatic and wall mounted type. The film is bathed automatically through three step: Developing, fixing and bathing. The time developing films needed can be selected in the designing range. During the bathing process, delivery mechanism adopts anti-flaming ABS engineering materials. It has easy cleanliness and maintenance, stable running, small volume, operators do not touch the developers and fixers directly, convenient and safe operation and so on characteristics.
2. Main Technical parameters
2. 1Normal working conditions
A). Surroundings temperature: 18º C-30º C;
(if the Room temperature lower than 18º C,it will Raise to work temperature automatically)
B). Relative humidity:≤70%;
C). Atmospheric pressure: 700hPa-1060hPa;
D). Voltage of electrical source: 110V or 220V± 20V;
about the power supply : we will send the power cord fitted buyer's country
E). Frequency of electrical source: 50Hz or 60Hz
2.2Surroundings of transportation and reservation
A). Surroundings temperature range: -40º C-+55º C;
B). Relative humidity range: ≤93%
C). Atmospheric pressure range: 500hPa-1060hPa.
(1). Processing time of film
The time which develops films needed is 2min-8min, and can be accommodated continuously. The error is within± 10%.
(2). Light leaking
During the bathing process, light leaking is not permitted.
(3)The stability of film delivery
The delivery mechanism of film should not lacerate or lock films.
(4)The timing panel of transit film
The timing panel of transit film adopts the number 1-12 represents month, each scale represents 10 days.
Weight of Unit: 7.5kg