BGD 186 is the newest Intelligent krebs Stormer viscometer which used to measure the paints and other coatings which indicate the viscosity by KU values. It is an upgraded and modified product of BGD 184 or STM-IV Stormer viscometer, through LCD show “KU” , “cp” , “g” value at the sample temperature at the same time. Meanwhile, it connects to the main part via a temperature probe, it can also show the temperature values of the measured samples.
BGD 186 is controlled by microprocessor, A paddle type spindle is drived at 200r/min by specific constant speed motor. The reaction torque of the spindle rotating at 200rpm is converted to a logarithmic function which display in KU (Krebs) Units by microprocessor. It is usually use for marking the viscosity of brush coating and roller coating. The instrument adopts special method of calculation ,it can delete data which deviated from the centre area automatically, it improved index of repeatability in measurement. The operator just need to let the spindle immerse into sample until a required depth. Then, they can read out the KU values or the CP value of the tested samples from the display screen directly.
The Stormer Viscometer adopts single slice microprocessor. A paddle type spindle is driven at 200rpm by a constant speed motor. The reaction torque of the spindle rotating at 200rpm is converted to viscosity in Kreb Units by microprocessor.
An operator immerse the spindle into the fluid. If the correct amount of fluid has been put into the container, the surface of the fluid will be at the immersion mark of the spindle. The operator can directly read out the KU values or the CP value of the tested samples from the display. This instrument are designed according with ASTM D 562 and GB 9269.
Main Technical Parameters:
1. Model No. : 186 ( New improved type based on 184)
2. Reference standard : ASTM D 562 and GB 9269 standard
3. Display KU value , temperature value and CP value of sample which is tested
By LCD screen together .
5. Measure range : 40.2KU~141.0KU(KU value); 27-5250 CP
6. Accuracy : ± 1.5% full scale
7. Reproducibility : ± 1% full scale
8. The magnetic basic plate is convenient to fixup sample receptacle
9. The portable rotor blade connector . The replacing and cleaning of rotor blade is convenient .
10. Rotor speed: 200r/min±0.5 r/min;
11. Volume of a container: approximately 500ml;
12. Lifting range: 90mm ;
13. Input voltage: 110~220V/50~60Hz ;
14. Power: 10W ;
15. Overall dimensions: 210mm×180mm×500mm (L×W×H)
16. Net Weight : 7.0KG
It used to apply in testing paint and other dote which showed KU value to identify Its viscosity . The operator just only immerge the mixing propeller blade into the stated depth of tested sample , then you can read the KU value and interrelated load value .