LCD Ammeter, LCD Ampere Meter, LCD Current Meter
1. Product Features 1.1 True RMS measurement 1.2 High-definition , large screen , dot-matrix LCD display s 3 phase voltage or current 1.3 Optional the upper/ lower limit alarm function, I / 0 switch functio n 1.4 Programmable, it may be arbitrar ily adjust ed the display ing value between 1 -9999 by the keys on panel 1.5 It ’ s Optional for communication function, communication address and baud rate .
2. Technical Index
Input Input Voltage:500V ( May customize for the special ) Current:5A , 1A ( May customize for the special ) Overload Last ing : 1.2 times I nstant: current: 2 times/second , voltage: 10 times/secon d Ex tension Extend measure range by instrument transformer or shunt Frequency 50Hz±10% Power Voltage AC220V±10% ( May customize for the special ) Power consumption ≤3W Display LCD Display time for the data updat ing :1s Display 1-9999 Precision Grade 0.5% Isolation W ithstand V oltage 2KV/50Hz/1min Insulation resistance ≥100M Ω Mean normal working time ≥9000h Work conditions Temperature:-25°C~55°C Humidity:≤93% No corrosion gas
3. Basic operation and useage 3.1 Display & Operation 3.2 Menu structure
3.3 Home Menu Page
3.4 The Device Menu of Set up
A fter e nter ing the “ SETUP ” page , through the cursor movemen t enter the goal to set submenu, then set the related parameter of the meter . For e xample: Press ing the ENTER button to enter into “system parameter” to set submenu, then we may adjust 3 phase ammeter (TA Current Transformer ratio ) or 3 phase volt meter (TV Voltage Transformer Ratio) and the mode of connection . Note: mode of connection "Y" represents 3 phase 4 wire, "the Delta " represent 3 phase 3 wire as below:
As shown above: 3 phase ammeter CT: 100/5A We should input 20A at the position of TA Ratio, then QUIT & SAVE.
4. Outline & Installing Dimension
Packaging & Shipping Package: 42*23.5*34 CM, 40pcs/CTN, Meter outline Size: 96*96 CM
My products are modularity structure, they may meet varies of demands.
We also provide customized serivce.
6. Attentions 6.1 The following conditions will lead to the device be damage or work abnormal: 6.1.1 The auxiliary power supply, voltage, frequency are over-range. 6.1.2 Please do not treat the PT signal as well as the auxiliary power supply at the same time Unless the PT have enough power, so that the meter can be ensured to work normally. 6.1.3 CT circuit current wiring terminal screws must be tightened, and ensure the inlet / outlet lines are reliable to avoid faults. 6.1.4 If checking meter, the calibration instrumentation should be prior a class 0.1 magnitude, so that the accuracy could be ensured. Skype: helen.fu16 Zhuzhou Sanda Electronic Manufacturing Co.,Ltd
Official Website: http://www.zzsanda.en.alibaba.com
Address: No.502, Wulidun Economic Development, Liling, Zhuzhou City, Hunan Province, China.