SA-160 antenna analyzers designed for testing, checking, tuning or repairing antennas and antenna feed lines. Mainly, these are S.W.R. (Standing Wave Ratio) and impedance measurement instruments (vector impedance analyzers). Easy-to use measurement modes. make SURECOM attractive for professionals and hobbyists. It is quite important that graphical display of various parameters over a wide frequency range is a key feature of these analyzers which significantly reduces the time required to adjust an antenna.
2.2 ” precision LCD screen,High-lumen LCD brightness, easy to read in direct sunlight Heavy duty Aluminum enclosure,strong anti-interference ability. Very compact and convenient for field use Built-in rechargeable Li-ion battery Full Band control (No need select Band ) Easy setup, only four control buttons with convenient on-screen soft menus Constant Battery Status Indicator and Auto Shut-off Built-in timer shows elapsed time when unit is turned on High efficiency IC, low power consumption, the longest working hours More choice measurement frequency Span . One botton quick scan mode HF1,HF2,HF3 or 3 band by full screen. Auto marker the Best s.w.r. level reference
BNC (50 OHM)
RF Output Power:
1-1.5V(5 - 10dB)
Build-in Battery :
3.7V 900mah ,Li-ion 14500
Voltage input :
Two-color LED Hidden (when not charging) there are three instructions state: 1, charging (red) ; 2, charging is completed (green) 3, Battery fault (flashing)
Standard Micro USB connector
70 x 120 x 25 mm
TFT LCD 320 x 240 pixels
Size 14500 of Lithium battery.
Smart charger, LED charging-status indicator
On-screen DVM, plus battery condition icon
Auto-off timer, Auto Run/Stop RFgenerator control
Control and Display
2.2 inch high-output color TFT
Function Keys :
4 botton, with on-screen (soft-menu) identifiers
Rotary encoder with press-in tuning-step selection
Weight and Dimensions
7 cm wide x12cm height H x 2.5 cm
6.1 oz (0.172 kg)
Turn on and off the power Turn on the power, Press and hold White botton when the dot disappear from left up angle ,. top Menu appeared on the screen. Turn off the power, You have 2 method : a)On Top Menu page ,Press and hold the White [F1 ] botton at 5 second . b)On Top Menu page , Rotary encoder on "Power off", Press the [F2 Enter]
1.1Function Select You can turn the knob select you want function and press [F2.Enter] botton .
2.0 ANTENNA TEST EXAMPLES by Scan mode 2.1[Scan] mode page In the Top Menu screen: Rotary encoder on "Scan", Press the [F2 Enter], to frequency mode measurement button, enter Scan (scan mode) screen. Press [F.2 scan ],button to start the scan operation. During scanning, in order to ensure measurement accuracy, each frequency will stay for short while. one scanning cycle will take about 1-2 seconds. Once SCAN is done, it will enter ”Present” mode (result) screen You have results on the screen : Show the SWR curve only (because default setting SWR curve only , Zx curve, Rs curve Disable) You can Enable other curve on the screen: On Top Menu page, Rotary encoder on "Zx", Press the [F2 Enter] to enable . Other cure will Disable when Power Off , If want enable in the future, Rotary encoder on "Save", Press the [F2 Enter] to save setting .
2.2 1. lnfull (default SCAN range 0.56MHz -59.9MHz,span 62.1MHz ) * Long press to [SCAN ]and press [F3 UPDn] select "1.Infull",and then release the scan botton. 2.lrnMk=(Learn marker move to center frequency) * Long press to [SCAN ]and press [F3 UPDn] select "2.IrnMk",and then release the scan botton. 2.3 [View] view the scan frequency point on the curve on each scan of the specific parameters of the measurement results. - Press the [F3.View ] button to start the view operation mode. rotary encoder ,the white triangle curve marker is move - Dot scale will display on screen, Press bottom change the Dot scale,You have 3 options of Dot scale mode for choose,[Dot =x1] [Dot=x10][Dot= x25] . (230KHz,2.3MHz,5.8MHz) SWR Full scale will display on screen,Press bottom change the graphically plot SWR,You have 3 options scale mode for choose, [SWR=x2 ][SWR= x 0.5][SWR=x1] .(1-9,1-19,1-4.5)
2.4[ Quick 1] start Scan mode: Quick view the scan frequency point on the curve on each scan Press the button [Quick 1 ] to start the Quick 1 operation mode .You have 3 options scan mode for choose . 1)Press the button [F1. Scan 1 ] 0.5 to 27.3MHz can of the specific parameters of the measurement results. 2)Press the button [F2. Scan 2 ] 16.6 to 43.4MHz can of the specific parameters of the measurement results. 3)Press the button[F3. Scan 3 ] 33.6 to 60.0MHz can of the specific parameters of the measurement results.
1 x SURECOM SA160 (ENGLISH VERSION) Colour Graphic Antenna Analyzer 1 x English User"s Manual 1 x Mirco Usb Charger Cable 1 x AC Adapter 100-240V (Depends on your country)
409 SHOP LTD is a manufacturing department of a professional enterprise providing two-way radio accessories. For years, we have been committed to developing products according to domestic and overseas markets' demands. Our full range of products include earphones, headsets, speakers, microphones, throat microphones, air acoustic w/mic, leather cases, batteries, chargers, programming cables, antennas and other fittings compatible with major brands of two-way radios.
We also provide customized solutions for customers depending on their requirements. Our products have been exported to Europe, America and Southeast Asia, enjoying high popularity.
We always persist in new and unique development and strictly follow the quality standard. Our materials are selected through selection process. As a leading department with innovative technology, we are proud of offering good quality products, low prices and better selections, as well as prompt and reliable services to our customers.