|Place of Origin:||Guangdong China (Mainland)||Brand Name:||samwa||Model Number:||YX-360RNA E-L-B|
|Measuring Current Range:||2.5mA,25mA,0.25A,*10A||Measuring Voltage Range:||
|Measuring Resistance Range:||X 1 ~2kΩ|
|Measuring Capacitance Range:||``||Measuring Inductance Range:||``||Operating Temperature:||0~40℃|
|Packaging Detail:||60 pieces per carton.|
|Delivery Detail:||3 days to 7 days|
1.capable of measuring DC voltage
2. AC voltage, DC current, DC resistance
3.Apply to schools, shops, factories for the family
YX360RNA E-L-B digital multimeter Specification
DC VOL TAGE
Accuracy at FSD :3,(1000v;5)
Extension:25kV(with HV probe extra)
AC VOL TAGE
Accuracy at FSD:4;(1000v;5)
Decibelmeter: —10to +22DdB
50μA(at 0.1VDC position),2.5mA,25mA,0.25A,*10A
Accuracy at FSD:3(10A;5)
X 1 ~ 0.2Ω up to 2kΩ,Midseale,at 20Ω
X 10~2Ω up to 20kΩ,Midseale,at 200Ω
X 100 X 100 ~ 20Ω up to 200kΩ,Midseale, at 2kΩ
X 1k~200Ω up to 2MΩ,Midseale,at 20kΩ
X 10k ~2kΩ, up to 20MΩMidseale,at 200kΩ
Accuracy at FSD:3
hFE 0 ~ 1000(with connector extra)
DCA range for # YX-360TRA-A
Ωrange for YX-360TRE;YX-360TRE-B;YX-360TRE-B-L
Set the range selector knob to BUZZ.,apply the test lead pins to two points to be tested and test continuity,Then the buzzer will buzz at between 0 and about 120Ω,It is the “LED”in the tester produce light,that mean the tested circuit is continuous.
Set the range selector to the “CONT’Y”position
Connect the test leads to the tested ciruit.
If the “LED” in the tester produce light,that mean the tested circuit is continuous.
Place the red test lead into the + socket and the black into the-COM
Place the red test lead into the DC 10A MAX socket and the black into the-COM
Set the range selector at a selector DCA range position.
Connect the red test lead to the positive polarity of the circuit tested and the black into the negative.
Read the DCV A scale converted with reference table.
ACV TEST ON OUTPUT TERMINAL
Plug the red test lead into the OUTPUT socket and the black one into the-COM.
Set the range selector at the selected range position.
Connect the test leads to the circuit to be tested and read the scale in the same manner as ACV test.Such a measurement is made to block the DC voltage which presents
In the same circuit and must be cut out so that AC voltage can be read alone.
ACV TEST ON OUTPUT TERMINAL
Plug the red test lead into the OUTPUT socket and the black one into the-COM
Set the range selector at the selected range position
Connect the test leads to the circuit to be tested and read the scale in the same manner as ACV test.Such a measurement is made to block the DC voltage which presents In the same circuit and must be cut so that AC voltage can be read alone.
Iceo(leakage current) test
For NPN transistor,the “N” terminal of the tester is connected with the COLLECTOR(C) of the transistor and the “P” terminal with the EMITTER(E) of the transistor.
For PNP transistor,reverse the NPN transistor connection.
For PNP transistor.A) connect the “N” terminal of the tester to the emitter of the transistor.B )plug the HFE connector into the “P” terminal and connect the clios in the same way as for NPN transistor connection.