II. Function Buttons and Instruction 1. ON/OFF(Switch on/off) Put batteries into the battery room of the scale, Place the scale on a solid and flat board without wind and vibration, keep the bubble in the center of the black ring. Put ON/OFF will show 88888 then 0.000, it’s ready to work. Press ON/OFF switch off the scale. If there is wind when weighing, please put down the wind shield. 2. UNIT(Unit switch button) l Press [UNIT] to switch weighing unit: g/oz/gn/ct/t/ozt l Press [UNIT] key for 4 seconds, all the units will light up and unit OZ blink, then press PCS to hold or close this unit, Press UNIT to select next unit, unit g can not be closed. Press [UNIT] for 4 seconds to return normal weighing state. 3. TARE Touch [TARE] to for Re-zero. 4. PCS(Counting function) Select 25pcs of samples(or 50,75,100pcs of samples) At normal weighing state, put the samples on the scale, then press PCS 4, seconds and the scale shows the weight of the samples, press PCS to confirm, then LCD will display PCS+ sample value. You can put more articles, the scale will show "PCS XXX", which is quantity of articles.
III. Calibration Function Turn on the scale, press ON/OFF key 4 seconds, the scale shows 0 then the weight value(if the capacity is 30g, 30.000 will blink), put 30g calibration weight on it, and shows PASS, then calibration is completed. (20g calibration weight is included with the product, you can press PCS to reduce the weight value to 20g, then put 20g weight to do the calibration), when the calibration value blinks, press UNIT to increase the value and PCS key to reduce the value. According to the amplitude of 10G to increase or reduce, we can select the right calibration weight, the system will remember the last setting operation.
IV. Auto power off setting Turn off the scale, press ON/OFF for 4 seconds and shows 88888 and F1-1, Press UNIT to choose the correct power on time(from 0 to 5), 0=do not power off automatically, 1-5 means the scale will power off automatically when no any operation on this time. You may choose the correct time as necessary. The scale can work normally after setting completed.
V. COUTION 1. This product is a precision measuring instrument. Please pay attention to operate, do not disassemble by yourself, if there is any problem, contact the after-sale person. 2. Place the scale on a solid and flat board without wind and vibration, keep the bubble in the center of the black ring 3. Use the scale in windless environment, Please put down the wind shield in windy. 4. Keep the scale out of your mobile over 50cm. 5. Please do the calibration when the weight environment changed, even if the temperature changes just 1 degree. 6. It will display “FULL” when overload. 7. Do not use the product in wet environment. Please remove the battery and keep it in a dry and cool environment when not in use. 8. Do not use corrosive cleaners to wipe the product.
SIMPLE OEM: MOQ 10pcs,print your company name on the label and instructions. OEM: MOQ 200pcs, also need pay USD150, after order quantity is over 200pcs, will refund the costs USD150. ODM: MOQ 1000pcs.
Q1: Are you a factory or trading company? A: We are a factory.
Q2: MOQ? A: We don't have MOQ limit, you can order any quantity you want. If you need scales with your specification, logo, packing, etc, please negotiate with us.
Q3:Can I have a sample for testing? A: Actually we think samples are not very important.You can buy from anywhere.But we think our cargos will better then any others. Samples will be 10%-20% higher than the cargo.
Q4:Can I add my logo on the machine? A:Yes,OEM is available for us. But you should send us the Trademark authorization letter.
Q5:How can i get the after-service? A:We will send you the spare parts by free if the problems caused by us. If it is the men-made problems,we also send the spare parts,but you should pay. For our range hood motors it is lifelong maintenance.
Q6:Do you have inspection procedures for these machine? A:100% self-inspection before packing
Q7: Delivery date? A: 1-3 days for sample, 15-20 days for 100pcs, 30-35days for 20HQ.
Q8: Payment Method? A: L/C at sight, T/T, Western Union and Paypal are all available. 30% deposit T/T in advance, rest against the Shipping Doc. 100% T/T in advance for sample order.
Q9: Shipping? A: a. International Experess: DHL/TNT/FEDEX/UPS (For sample) b. By air (For sample orders.) c. By sea (15-45days), port of loading: Shanghai d. By domestic logistics to your forwarder. (2-3days)
Q10:Can I have a visit to your factory before the order? A: Sure,welcome to visit our factory. We can pick you up in the airport or train station. Hotel booking service is available