Hospital Medical Anesthesia Equipment, Anestesia Machine For
What is the detail of our hospital anesthesia equipment?
Prospective service and support
MCA-850 Anesthesia Machine is equipped with an accurate dedicated anaesthetic vaporizer and a safety device for preventing cyanosis and necessary alarm system. During the anesthesia, the patient’s respiratory functions can be managed by using a microcomputer controlled pneumatic electrically controlled synchronizing anesthesia respirator. Each connection part of the whole machine is a standard interface. A highly efficient and large volume soda lime absorber can minimize the re-inhalation of carbon dioxide by the patient.
Power On Self Test
The automatic conversion for standby power while AC power outage, keep working not less than 1 hour.
Na Lime Tank: Equipped with international advanced one-piece soda lime cans,it can be fast replaced with hands and withstand temperatures up to 134℃
Circuit: Metal integrated breathing circuit, Perfect function of gas sealing under high temperature disinfection, and easy to cleaning and disinfected.
Flowmeter: 5-Tubes flowmeter;
1. Assure that O2 output not lower than 20%
2. NO2 automatically off while O2deficiency
Display & Operating Interface:
10.4'' LCD Display Screen, can feedback patient data
Shuttle button: More accurate
Vaporizers: En flurane/ Iso flurane/ Sevo flurane(Option: Halothane)
Setting Range: 0.5~5%
What is the specification of our anesthesia machine?
10.4 '' LCD Display Screen
Adult & Child
pneumatically powered and electrically controlled system
Closed; Semi-Closed; Semi-Open
Breathing circuit integrated standards
5 Tubes Flowmeters: O2:0.1~10L/Min, N2O:0.1~10L/Min; Air: 0.1~10L / Min
Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and easy maneuverability with foot operated brake provisions
TWO drawers totally extractable
Medical oxygen and nitrous oxide with a pressure ranging from O2: 0.32~0.6MPa ; NO2: 0.32 MPa to 0.6 MPa.
one deep breath per 70 ~ 120 controlled respirations, the inspiration time is 1.5 times of the setting point
Maximum safety pressure:
≤ 12.5 kPa
Pressure limit range:
0 ~ 6.0 kPa
Airway pressure alarm: Audible and visual and with yellow and red color indicating
Lower: 0.2kPa~5.0kPa; Upper: 0.3~6.0 kPa
Tidal volume alarm: Audible and visual and with yellow and red color indicating
upper alarm: 50 to 2000ml, lower alarm: 0~1800ml
Oxygen concentration alarm: Audible and visual and with yellow and red color indicating
upper alarm: 21% ~100%; lower alarm:10%~80%
Power Supply Alarm
Ac/dc power supply are after failing to send out alarm immediately Alarm Time: keep >120s
The airway pressure continues to be higher than 15 hPa ±1 hPa for 15s±1s, then the machine will raise an audible alarm, the pressure will be displayed in red and the continuous high pressure red alarm wording is displayed on the screen of the anesthetic respirator.
10 ~ 40
no higher than 80 %
860 hPa ~ 1060 hPa
100-120 Vac, 50/60 Hz;
Attention: the AC power supply used for the anesthesia machine must be well grounded.
Attention: the anesthesia machine used must be equipped with a carbon dioxide monitor complying with ISO 9918:1993, an oxygen monitor complying with ISO 7767:1997 and an expiratory gas volume monitor complying with 220.127.116.11 of Medical Electrical Equipment Part II: Special Requirements for Safety and Basic Performance of Anesthesia System.
-15 ~ +50
no higher than 95 %
86 kPa ~ 106 kPa.
It should be stored in a room without corrosive gas and well ventilated
comply with the requirement of GB/T 15464
Between the packaging box and product, soft material with appropriate thickness provided to prevent loosening and mutual friction during transportation
Moisture protection and rain protection to ensure that product is protected from natural damage.
Safty & Alarm
It alarms when the oxygen supply from pipe or cylinders lower than 0.2MPa
Ventilation Volume Alarm
Lower: 0~12L/Min; Up: 18L/ Min
It Alrmas while AC and DC supply failure ; Keep alraming time: >120s