Inner package------ Standard carton with foam.
35 days after payment
Size:φ 5*330mm/10*φ 330mm
Trademark:Geyi/Euprun Origin:Zhejiang, China
Dissecting Forceps (Celiac separation clamp)
I. Purpose In cooperation with the laparoscope, celiac separation clamp is used with abdominal puncture machine whose purpose is to separate the adhesion organization in the operation. It applies to Department of obstetrics and gynecology, general surgery clinical and family planning clinics.
II, Type Separation clamp is divided into double joint and single joint, and both outside diameter is φ5.0mm. According to the head of the bending Angle, they are divided into:double joint:0°Separation clamp,30°Separation clamp,45°Separation clamp single joint:0°Separation clamp,30°Separation clamp,45°Separation clamp Model Head end diameter(φ) working length(L) 0° double joint 30° 45° 5.0 mm 310~350 mm 0° single joint 30° 45°
III,Main technical parameters 3.1 Head end diameter of separation clamp is φ5.0mm ; working length is 310 ~ 350mm;opening angle of double joint is not less than 45°,opening angle of single joint is not less than 30°; 3.2 The head end tooth shape should mesh when the separation clamp closes. 3.3 Separation clamp can rotate 360 degree all directions, it can turn convenient, and it shouldn't be stuck. 3.4 The clean-out hatch of the separation clamp outs water flowing, it shouldn't be clogged. 3.5 Separation clamp should have a good clamping force. 3.6 The clamp head (inner core) of the separation clamp and the handle are demountable type connection, and it install conveniently. 3.7 The head hardness of separation clamp is not less than 330HV0.2. 3.8 Separation clamps should have good corrosion resistance and required surface roughness.
IV,How to use ?
1.The clinician who is trained qualified and familiar with this equipment structure performance use this surgical instrument according to the code requirement.
2 In a direct laparoscopic, the separation clamp (head should be in closed state) which has the same external diameter with the abdominal puncture machine is plugged into the puncture clamps slowly through the setting of abdominal puncture machine.
3 In a direct laparoscopic, put the separate clamp into the operation area, and then handle independent or cooperated with other machine after the clinician is sure the operation area is right.
4 If the clinician finds that the head style of the separation clamp is not suit with the operation area in practice, he should replace the right separation clamp.
5 The clinician should keep the vision clearly in practice and make sure the operation is done successful.
6 The separation clamp should be kept by special person, and it should be cleaned and disinfected before and after using. And you can use the vacuum pressure steam sterilization. When you are cleaning, you should remove the clamp head(inner core) and the handle. And after disinfecting the separation clamp, you had better use deionized water washing. The separation clamp should be dried after washing immediately.